This doesn't bode well for those that need the 64-bit version, though. 0 Kudos Message 7 of 8 (1,090 Views) Reply 0 Kudos Re: Error -229771 at DAQmx Start Task.vi GurraW In particular: 1)Why in the "resample VI" there is the need to specify the number samples per cycle and what does it change if I change this number? 2)Why do i think there are no other solutions. Show more post info Size: 360 bytes Customize: Thread profile page for "ni max import error" on http://www.ni.com. Now I have two problems. Source
Please contact National Instruments Support." Recurring Error "Internal Software Error occurred in MIG software. More site info... Seems pretty pointless. As for the errors that occured, nothing hinted at a database issue, there were absolutely no issues in MAX, I was able to delete and recreate the one task we were
But, LV didn't actually install, because the installer thought it was already there. You could take this a step further: place a single array control, containing the cluster and sized to show a single element, on your front panel. Just use one and branch the wire. Applications Engineer National Instruments 0 Kudos Message 8 of 12 (4,101 Views) Reply 0 Kudos Re: Error – 229771 occurred at DAQmx Start Task TimA2011 Member 01-11-2013 04:07 PM Options Mark
United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Discussion Forums : Most Active Software Boards : LabVIEW : Error -229771 at DAQmx Start Task.vi LabVIEW Register NI Tech support walked me through all of the DAQmx Corruption Correction procedures and we reloaded LabView 2009 and all DAQmx drivers. I should get 3 points in the Spur List. Thanks, hiNi0 0 07/28/15--09:05: Re: C library function call - Unavailable Type for one of the parameters in the Ignoring that issue for now, I went ahead and attached two examples that I would try: one uses the on-board clock of the DAQ device and the other is software-timed.Best Regards,
The simplest example of a symptom is to create new blank VI, drop a DAQmx function on there, e.g. Internal Software Error Occurred In Mig Software It is good, in that it describes the general steps that are involved, but it is only covers up to XP I think. vcb88 Solved: Error Code -229771 - Discussion Forums - National Instruments error -229771 DAQmx Create Channel (AI-Voltage-Basic) Internal ... http://forums.ni.com/t5/LabVIEW/Error-Code-229771/td-p/1211485 Which version of LabVIEW? 3.
Please contact National Instruments Support." So I asked (myself), "What is MIG Software?". All rights reserved. The reason it's not closing properly right now is because you're not passing the correct value, so it doesn't properly deallocate the structure.0 0 07/28/15--09:09: Re: one channel control This thread profile page shows the thread statistics for: Total Authors, Total Thread Posts, and Thread Activity Home| About Us| Submit Your Site| Update Your Site| Get Search For Your Site|
What OS are you running?2. http://126.96.36.199/t5/LabVIEW/quot-Internal-Software-Error-occurred-in-MIG-software-Please/td-p/2368538 Why do you have duplicate diagram constants with the same value (e.g. Daqmx Error Codes After reading the Knowledgebase and other people's comments, I am getting by the error (very catastrophic as nothing will work) by the following: 1. Seems too much!
Please contact National Instruments Support." Recurring Error Luke_W Member 04-04-2013 02:41 PM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a this contact form I am using Windows Vista, LabVIEW 2009 SP1, NI-DAQmx 9.0.2, HP laptop withMXI-4 interface to chassis PXIe-1073, DAQ hardware NI PXIe-6366. If I run the vi outside of the project (just open the vi), I DO NOT get the error. At http://www.ni.com/downloads/instrument-drivers/ you will find instructions on what a driver is and how to use one.
I have completely rebuilt my system, from a drive reformat, only to have the problem return. Before I uninstall and re-install NI-DAQmx as a last resort, I am hoping this is something someone has come across before.Note also that this appeared for no apparent reason. what i was trying to do is that i want this vi to show me the output signal of the analog comparison event which carries the information about the times that my http://robertwindows.com/labview-error/labview-error-codes-1097.html There are also numerous free tutorials on LabVIEW itself. 0 0 07/28/15--10:53: How to use a event structure in a subVI Contact us about this article I am trying to use
The real current signal must be read by a Rigol Oscilloscope, and then I have to sample the magnets with the relation between current and magnetism. This is what I When I first start LabVIEW after a cold boot, I open a small test VI I created, which contains only the DAQmx Create (Voltage) Channel function. pr65.tiredwatch.xyz/c38l_fix-jwrc-dll-error.html Norton Internet Security - Wikipedia, the free encyclopedia Norton Internet Security, developed by Symantec Corporation, is a computer program that ...
NI MAX - error message "There was a problem updating the settings for this device. I'm told I'm using the latest versions of Labview 64-bit version and DAQmx, which were obtained and installed less than a month ago. Yup, else you would spell it LabVIEW jpc335 wrote: Example String of Hex Values 03 23 F6 21 B4 C4 74 F6 The first sensor requires bits 3 &4 (21 Poor|Excellent Yes No Document Quality?
Contact us about this article Dave- Thanks for the reply. If I attempt to not supply a username/password, the Email Send VI just times out, evidently because it can't connect Service mxssvr is running Process nixms.exe is not running NOTE: This error only occurs while running from Project Explorer. Which Version of DAQmx are you using? 9.0.2 is the most recent version (an upgrade/repair would be a good troubleshooting step in this case). Check This Out This error can be caused by the Measurement & Automation Explorer's database becoming corrupt.
If you go to Microsoft.com you can download Microsoft Security Essentials for free. Were there any significant changes that happened on the machine around the time it started to misbehave (software install/update, hardware change, other crash, etc) or has it always been an issue? Thank You, RonC Solved! Please Contact NI for all product and support inquiries.
I am also using a NI cDAQ-9174, with the first 2 bays populated with a NI 9205 and a NI 9401. Contact us about this article OK so im a mechanical guy, not a programming guy so lets start there. I just learned how hexadecimals work and have limited experience in LABview. I don't have access to the host machine right now, so I can't give you the exact numbers. Check for NI Configuration Manager and NI Device Loader: This error could also be caused by having installed DAQmx from an account that is not a system administrator.
I have the USB 6008 wired for a loop back test.Any other ideas?Thanks,Bryan 0 Kudos Message 3 of 12 (6,203 Views) Reply 0 Kudos Re: Error – 229771 occurred at DAQmx Attachments: Report Date: 09/20/2010 Last Updated: 11/06/2015 Document ID: 5DJC03WW Your Feedback! I need to find the file that is telling the installer that there is stuff installed, even after I have removed it, I think, to get rid of the corrupted install.