Either the hardware between the source and destination terminals does not support the inversion, or other routes in the task might be interfering with this route. Turn the computer off and on again. The position and offset specified a sample prior to the first sample acquired (sample 0). Also, verify that the rate of the external clock matches the specified clock rate. Source
Error -200057Duplicate device identifier in the device list. For lower sample clock rates, use an external sample clock or an external sample clock timebase. Change the marker position or positions to fit within the waveform, or increase the size of the waveform. Refer to the documentation for details about settling time constraints.
Error -200135Counter timebase source and counter timebase master timebase divisor settings are inconsistent with one another. Error -200250Length of waveform is too small for the last "generate" statement in a "repeat until" loop. Now, with the SensorDAQ you want to use connected, add a new DAQ Assistant block to the diagram. Error -200248Specified marker position is too close to the end of the last "generate" statement in a "repeat until" loop.
Error -200046External calibration section of the EEPROM on the device appears to be corrupt. Register for an account your email your username A password will be e-mailed to you. Error -200004Input termination resistor for one or more channels was overloaded. Remove all of the channels currently in the task except the channel that will be used as the analog trigger source, or change the analog trigger source to a terminal.
Error -200243Hardware clocking error occurred. Contact National Instruments if the accessory appears to be functioning improperly. Related Links: Drivers and Updates: DriversKnowledgeBase2HD9BP6I: My Devices Are Not Listed in Measurement & Automation Explorer (MAX, but They Show Up in Windows Device Manager Attachments: Report Date: 06/17/2008 Last Updated: Error -200153Reference voltage applied for calibration is outside the range defined for calibration of this device.
Increase the period between ADC conversions. For slower rates, you must use an external convert clock or an external convert clock timebase. Error -200125Lines 4 through 7 of this port are configured for output. Error -200042Inversion requested is not possible.
Stop other tasks that are connected to this device. 2. http://www.techyv.com/questions/error-201003-labview-project To remove this warning, slow down the sample clock, or else change your application so that it can keep up with the sample clock. Use the handle obtained when the calibration session for this device was originally opened. If you are generating your clock internally, please contact National Instruments Technical Support.
Sign Up Thank you for viewing the Vernier website. http://robertwindows.com/labview-error/labview-error-1445.html Also, verify that the rate of the external clock matches the specified clock rate. From the menu window that opens up select the current device configuration and confirm that it is the right one by checking inMAX to see that the device number matches. Password recovery Recover your password your email Search Sign in Welcome!
Error -200040Source terminal to be routed could not be found on the device. Error -200009Counter 1 DMA acquisition started while starting, committing, stopping, or uncommitting an analog output (AO) task. Rewire the block diagram. have a peek here Error -200220Device identifier is invalid.
A couple weeks later the LabView program no longer works, instead giving Error 201003. After the module shows up in Device Manager, go to Measurement& Automation Explorer (MAX)and check if the device is visible under Devices and Interfaces. Insert an appropriate integer at this location in the script.
Error -200112Calibration handle is invalid. If the second computer recognizes the card, but the first doesn't, then there is something wrong with the first computer, either bad/incompatible PCI slots, motherboard, or other hardware.If neither computer recognizes Refer to your documentation for details. Error -200012Clock rate specified exceeds the maximum conversion rate of the ADC.
The following constraint must hold: Offset < (Gain * Maximum Value / 2) Error -200270Interpolation rate specified is not possible for the given sample rate. Identifiers cannot start with a number. Consider settting either the sample clock rate or the sample clock divisor, but not both. http://robertwindows.com/labview-error/labview-error-6.html Error -200120Port is configured for input.
Error -200155Reference resistance applied for calibration is outside the range defined for calibration of this device. Error -200054EEPROM of the device appears to be corrupt. Error -200141Data was overwritten before it could be read by the system. Error -200022Attempted writing a sample value that was too small.
Find My Dealer Prices shown are valid only for International. Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase English 1 ratings: 3 out of 5   Error 201003 at Test Panel in If Data Transfer Mechanism is Interrupts, try using DMA or USB Bulk. What I have found is that my problem occurs when I try to run a task without the task started. I have restructured my code...
Error -200056The connection count of the attached accessory has exceeded the recommended limit. Error -200091Shared library version installed is incorrect. Error -200026Waveform name was expected, but not found in the script. Error -200149External timebase rate must be specified to translate the delay into ticks.
The device will shut down if its temperature exceeds the maximum recommended operating temperature. If you are using an external clock, check your signal for the presence of noise or glitches. Error -200045EEPROM of the device appears to be corrupt. The conflicting properties must satisfy the following constraints: Maximum <= Counter Timebase Rate / 2 Minimum >= Counter Timebase Rate / Counter Maximum Count.
Error -200009Timestamps have been overwritten. Set the Counter Timebase Rate property to the appropriate value for your external source.